# Elekta Synergy S Linear Accelerator Lab Analysis

Good Essays
Elekta Synergy S linear accelerator (LINAC) was used to produce the 6, 10, and 15MV photon beams for measurements. The accelerator is fitted with the new Agility multileaf collimators (MLC) thinner than the previously used MLC’s to improve a field definition (Kantz et al. 2015; Thompson et al. 2014). The quality assurance was carried out monthly to ensure the LINAC response does not vary from the commissioning data (CPQR 2013). The LINAC has been calibrated to 100 cGy/MU at Dmax for a 10×10 cm2, and daily output checking was performed to ensure that LINAC operated within the ±3% limit (Andreo et al. 2006). 3.2 Film
3.2.1 Kodak X-OMAT V/EDR2 film
3.2.2.1 Calibration curve
The radiographic EDR2 films which were used for investigation of PDD,
The right image shows the sensitive volume of a diode or diamond detector
The detectors were aligned visually with the cross hairs (centre of the beam, CAX), and then the OmniPro software is used to correct the positioning of the detector. The servo measured a depth dose curve to find the position of Dmax and 40 % of a maximum dose. Then in-line and cross-lines profiles are measured at these depths, and the position of the 50% level was determined for these profiles. The positions of the 50% for the profiles are compared and the offsets are then displayed.
3.3.2.2 Voltage
Diode and micro-diamond detectors are semiconductor detectors thus no voltage is applied, CC01 required 300 volts to allow for optimal data
The two detector method employs the use of a field and a reference detector. The reference detector is used to minimize the effect of dose rate fluctuations (Das et al. 2008). The single detector method utilizes the step-by-step mode of data collection, this mode moves the detector at specified step sizes and stops for a specified time. This minimizes the linac dose rate fluctuation which is the same function performed by the reference detector. The pause time was determined by comparing the profiles obtained from two detector method and the single detector method to optimize both the beam on time and the noise.
The single detector mode is further optimized by utilizing two algorithms that are within the Omnipro software. The gradient algorithm is used in measuring beam profiles as the detector’s step sizes alter if two positions have a difference that is more than predicted. The breakpoint algorithm was used for PDD measurements, as the algorithm changed the step sizes after a certain depth was reached. The output factors were measured by collecting signals with the sensitive volume of the detector placed at 10cm

## Related Documents

• Decent Essays

The visible pulse (gating pulse) enters the microscope, and together with the ultrafast electron pulses illuminate the nanostructure specimen (gold nanoparticle). The visible laser pulse kept at low power (~1.8 mJ/cm2) to avoid saturation. At the spatiotemporal overlapping, the coupling between the visible and electron pulses takes place. The signature of this coupling can be revealed by measuring the electron energy spectrum using the electron energy spectrometer attached to the microscope. The measured electron spectrum is shown in blue line curve in Fig.…

• 1722 Words
• 7 Pages
Decent Essays
• Decent Essays

The heat released by the process can be explained by a rate equation. It is generally an assumption that the absorption of the photon is expressed by a linear relaxation process, so the heat released can be determined by solving the rate equation. The heat released can be delayed by some gases mixtures when the excess of energy is somehow taken out by collisions to a long lifetime of other molecules. In cases of very short laser pulses there is a possibility of optical saturation, in that case the heat released is no more a linear function of the intensity of light and the absorption coefficient. Fig 1: Elementary process occurring during generation of PA…

• 931 Words
• 4 Pages
Decent Essays
• Decent Essays

8.3.2 Linear calibration using a least squares regression can also be used. The “goodness of fit” of the linear regression, given as r2 by the software, must be greater than or equal to 0.995. 8.4 Verify the calibration curve with an independent standard for each gas. The second source ICV must be within 20% of its expected value for each analyte. 8.5 Also examine the calculated value of the lowest calibration standard (MRL).…

• 1252 Words
• 5 Pages
Decent Essays
• Decent Essays

Detailed interpretations of the infrared and Raman spectra of 2NAP are reported. The theoretical spectrograms for FT-IR spectra of the title molecule have been constructed. The 1H and 13C nuclear magnetic resonance (NMR) chemical shift of the molecules are calculated by the gauge independent atomic orbital (GIAO) method and compared with experimental results. Further, density functional theory (DFT) combined with quantum chemical calculations were done to find out the first-order hyperpolarizability. The calculated HOMO and LUMO energies shows that charge transfer occur within the molecule.…

• 2780 Words
• 12 Pages
Decent Essays
• Decent Essays

The structure information can be determined by analyzing the peak of diffraction angle 2θ degree. X-ray diffractometers consist of three basic elements: an X-ray tube, a sample holder, and an X-ray detector. The different annealing time will cause different microstructure, and people can summarize the change rule of the annealing time vs. structure. Result and Discussion Thicknesses measurement data Group 1: 12156.525 Å Group 3: 15074.94…

• 916 Words
• 4 Pages
Decent Essays
• Decent Essays

One can see that the peak of the distribution for all channels has come closer to zero after applying the RARMA algorithm. This figure should be accompanied with Table 2, which tabulates the resulting mean radiance departures before and after applying the radiometric adjustment procedure. Computed biases seem to vary from one channel to another, which is to be expected. Up to ~5 K bias is observed in some channels, more specifically in channels 3 and 4 (18 GHz). Nevertheless, after applying the RARMA algorithm, a significant bias reduction has been possible.…

• 881 Words
• 4 Pages
Decent Essays
• Decent Essays

The overall objective of this lab was to verify Bernoulli-Euler beam theory for the lip-loaded cantilever and use that to calculate the Poisson ratio for the material in this case 6061-T6 aluminum. This was done by having to different objectives for each part of the Lab. Part one was use electrical resistance strain gage, this was to measure the strain a cantilever beam as it undergoes a known deflection. We would then compare the measurements with that of the strains predicted by simple beam theory [1]. The objective for part two was to use two electrical resistance strain gages on the cantilevered beam to measure and calculate the Poisson ratio of the beam.…

• 2086 Words
• 9 Pages
Decent Essays
• Decent Essays

The correlation between prompt LII and soot volume fraction obtained by laser extinction is investigated. The results are discussed in detail at the end of the chapter. Chapter \ref{ch:Inertgas_Results} is dedicated to the investigation of the mitigation of soot inception by adding $N_2$, $Ar$ and $CO_2$ separately to the fuel supply. For this, laser edge extinction and slice method are introduced. Planar elastic light scattering (PELS), TiRe-LII, and two-color laser-induced incandescence…

• 794 Words
• 4 Pages
Decent Essays
• Decent Essays

This X-ray spectrum consists of several components. These X-rays are then collimated to get Kα1 and Kα2 radiations then directed towards the sample. When Bragg’s law is satisfied, the constructive interference occurs and its corresponding intensity peak is observed. In the θ-2θ mode spectrometer, X-ray source remains in fixed position and the detector fixed in the arm moves to collect the diffraction intensities. The intensity of the X-rays is recorded with the help of the detector and process the signal to a count rate.…

• 1032 Words
• 5 Pages
Decent Essays
• Decent Essays

First part is an observing and recording of fluid flowing out from a test tube and in second part a Venturi apparatus will be used. In both cases Bernoulli’s equation can be simplified to calculate the velocity of fluid according to experimental conditions. In the first part of experiment, the pressure is the same for both ends which allow us to remove the pressure and gives: ρgh₁ + ½ρv₁² = ρgh₂ + ½ρv₂² This can be simplified further by assuming that the velocity at the surface is 0 as it is moving very slowly and the height of the bottom is 0. Using this assumption gives ρgh₁ = ½ρv₂² Since it is the same fluid so the density, ρ, is the same which they can be cancelled out to give ghsurface = ½…

• 928 Words
• 4 Pages
Decent Essays